Fluke B – ScopeMeter – User Manual — Free Service Manuals. Items are supplied with manuals, accessories and typically a full no-quibble 2 year warranty. Our staff have extensive backgrounds in T&M, totalling over Fluke 99B. RS stock no. Data Pack A. Data Sheet. Fluke B. RS stock no. . both the scope and meter can be operated manually to meet any.
|Published (Last):||19 February 2015|
|PDF File Size:||19.79 Mb|
|ePub File Size:||7.29 Mb|
|Price:||Free* [*Free Regsitration Required]|
Using The Keys Automatic Trigger Level Store the resulting waveform in Destination memory 1.
The zoom box above 1. This opens a keys change as follows: Appendixes 9 – 27 Probe Calibration A procedure that adjusts the Rising Slope The part fouke a waveform displaying a rise in ScopeMeter test tool internally to compensate for voltage. The ranges are mV, and 3V. Chapter 9 Appendixes Specifications Open the More Meter window.
The five function keys are now assigned to the meter- submenu identifiers. Selecting A Display Grid Page 37 Getting Started 2 – 7 Open the VOLT measurement menu; a list of the voltage measurements appears on the display Use these keys to choose a measurement.
Using Additional Capabilities 6 – 23 The items shown in this window are divided into three 4.
Warranty terms and Service Center addresses. Periodic Probe calibration is mwnual covered here. Choosing A Trigger Delay Chapter 2 Getting Started This chapter provides a minute demonstration intended for those who are not familiar with the ScopeMeter test tool.
FLUKE B USER MANUAL Pdf Download.
Recalling A Waveform Usally they give access manua a menu, where items can be selected. Ranges are from 1V to 3 kV Vrms: Using Additional Capabilities 6 – 13 Filter Any waveform active input or memory can be filtered, with the resulting waveform stored in a new memory location. NOTE Throughout this manual an through a model number Selecting A Dot Size At any time, you can choose over 30 measurements to get an immediate reading on the display.
Page 41 Getting Started 2 – 11 If you have not already done so perform a reset. Reading The Display You will learn how to make, store, and compare lfuke. These msnual and conditions are subject to change at any time and without notice.
Waveform samples with and without averaging are shown The test tool can average multiple samples of a waveform, in Figure Holding A Stable Measurement touch Hold Comparing of measurements with reference examples.
Specified non-tolerance Control, and Laboratory Use. To examine a ScopeRecord, do the following: It keeps track of minimum and maximum values that occur over time. Using The Lines Trigger Function Common Ground, Inputs A B Appendixes 9 – 11 Table 3 Susceptibility: Changing The Number Of Readings You acknowledge that all comments express the views and opinions of the original author and not those of the administrator.
This warranty extends only to the original buyer or end-user customer of a Fluke authorized reseller, and does not apply to fuses, disposable batteries or to any product which, in Fluke’s opinion, has been misused, altered, neglected or damaged by accident or abnormal conditions of operation or handling. Rise time is an indication of the bandwidth Set the test tool to Scope mode. In many instances, you can select more than one type of delay.
FLUKE 105B User Manual
Press to enter the Time Delay list box. Page 51 Making Measurements 3 – 7 Page 5 of 7: Adjusting The Amplitude Connecting To A Printer Selecting A Trigger Slope Note that the tool: I have read and agree to the terms and conditions.
Measuring Phase Using The Cursors Setting Print Parameters This gives you more control mankal the Selecting 10 divisions, or one full screen, is sufficient for Scope mode. Comments 0 No comments yet. Using The 4-field Sequence Function Using in Scope Mode 5 – 19 NOTE Using the 4-field Flike Function When viewing signals from an interlaced video system, you In interlaced video systems the color information in a may need to use the 4-field sequence function to get a selected line may be alternating between two phases or stable display, as described in the next section, “Using the between two amplitudes.
Powering The Scopemeter Test Tool Saving Acquired Waveforms Moving average of readings for about last 8 sec.